Important Safety Information
For a complete list of cautions and contraindications please review the
TubeClear Operator’s Manual.
For further Indications for Use, Contraindications, and Cautions, please review the
TubeClear G Model Directions for Use.
For use with TubeClear Control Box Model 101 and Clearing Stem Models G-1008, G-1009, G-1010, G-1011, G-1012, and G-1014.
G Clearing Stem Contraindications
Do NOT use the TubeClear System if:
+ You do NOT know for certain the size (French and length) and manufacturer of the Feeding and Decompression Tube.
+ The G TubeClear Clearing Stem Model is NOT available for the type, size, and length of the Feeding and Decompression Tube.
+ The G Feeding and Decompression Tube has been removed from the Patient.
+ The G Feeding and Decompression Tube requires placement or repositioning.
+ The G Feeding and Decompression Tube has been in use for longer than the Tube Manufacturer’s recommendation.
+ The Patient is not permitted to receive liquids.
G Clearing Stem Cautions
Make certain that the correct Clearing Stem Model has been selected for type, size, and length of the Feeding and Decompression Tube.
+ Use of the wrong Clearing Stem Model may result in over insertion. Over insertion of the TubeClear Clearing Stem may cause harm to the Patient’s stomach or intestines.
+ Use of a Clearing Stem in a smaller French size Tube than indicated may result in the Clearing Stem becoming lodged in the Tube, and possible Tube dislocation when the Clearing Stem is removed.
The TubeClear System contains magnets. Magnets are inside the Control Box and at the end of the Clearing Stem that attaches to the Control Box. Do NOT place the Control Box and the Magnet at the end of the Clearing Stem within 15 cm (6 in) of pacemakers, defibrillators, and other electronic devices (including, but not limited to, credit cards, video tapes, televisions, computer monitors, and other CRT displays) because the Magnets may interfere with the operation of those devices.
Federal law (U.S.) restricts the TubeClear System to sale by or on the order of a physician.
The TubeClear System is intended for use by Certified or Licensed Healthcare Practitioners. Reading the Operator’s Manual is required before using the TubeClear System. For information about additional training, contact Actuated Medical’s Customer Service Department at +1 (814) 355-0003 ext. 117 or visit the website TubeClear.com.